Portable Pressure Steam Sterilizer Precautions
1, before each use, check the main body have enough water 4 liters, water level exceeds the combined heating pipes.
2, at the start of heating, venting valve wrench should be allocated to "blowing off steam" position. Sterilization with heating barrel so that cold air from escaping. Otherwise, steam penetration will be hindered air sterilization effect and can not be expected, put the steam valve after the exhaust, put down the wrench, such as the emergence of a large number of vapor leakage, simply pull the lever up and down repeatedly a few can.
3, sterilization liquid, the liquid should be filled within a rigid heat-resistant glass, liquid volume should not exceed the volume of bottles 3/4. Bottle with a cotton yarn Buser well wrapped and tied in with Shasheng bottleneck. Do not use perforated rubber stoppers and cork, to prevent the glass burst. Glass is best to put it slightly larger than the inner volume of enamel or metal disk, in case of glass burst, fluid loss and loss Buzhi Yu and the inner wall of the sterilizer. Pay particular attention to the end of the sterilization, do not immediately release the steam, in order to avoid accidents. To be in the gauge pointer to zero before they can open the cover removed sterilized liquid.
4, different types, different sterilization requirements of items such as dressings and liquid, etc., do not put together sterilization, in order to avoid a loss, loss.
5, the end of the sterilization, if the pressure gauge has returned to zero, but not open the cover. This time should be allocated to the bleed valve wrench "blowing off steam" position. So that outside air into the sterilizer, to open the cover.
6, after prolonged use pressure gauge, if the pressure gauge pointer can not reply to zero, should be timely maintenance. Usually should be regularly with the standard gauges to check, if not normal, the table should be renewed.
7, plus the best sterilizer ionized or distilled water, thus reducing water and chemical precipitation of solutes during the sterilization, deposition and electric sterilizer bottom surface. Effect of heating pipe work properly.
8, after prolonged use rubber seals of aging, should be changed regularly.
9, when the sterilizer is not used, should be kept clean and dry sterilizer. This extends the service life.
10. After the sterilization process, to open the cover, pressure gauge pointer must be in the zero position.
Next：About Shanghai Shen An instrumentarija "three cards one" change of heading notice